Hi, I have one situation that for me is complicated. I run CAT6 UTP network cable. When I connect each end of the cable to a Fluke DSP-4000 & DTX-1800 Network Analyzer, it tells me that Return Loss failed (RL and RL @ Remote). I check details on the analyzer it says that pair 3,6 failed 4,5 failed and the others passed. All the cables are inside of metal gutter. I have 2 questions:
What kind of problem can give Return Loss failed (RL and RL @ Remote)?
I just want to be sure that this network will work in 1000 MB.
Free Guide: Managing storage for virtual environments
Complete a brief survey to get a complimentary 70-page whitepaper featuring the best methods and solutions for your virtual environment, as well as hypervisor-specific management advice from TechTarget experts. Don’t miss out on this exclusive content!
No problem! Submit your e-mail address below. We'll send you an e-mail containing your password.
Your password has been sent to:email@example.com
To follow this tag...
Thanks! We'll email you when relevant content is added and updated.
Share this item with your network: